The physical characteristics of CeF3 are well suitable for detection of domestic pulse γrays. The time response of detector composed by phototube with CeF3 are measured by use of the pulse radiation source with rise time about 08 ns, and FWHM time 1.5～22 ns.
With position length of greater than 3m, sensitivity of better than 25μm, and nonlinear error of 0. 6% in linear range of±1. 5mm, this positioner can generate scanning 1aser beams using a PZTA, which are de-tected by a two-quadrant photocell, and then the aberration value of the signal is output-ed digitally after the photocurrent signal is processed by a monolithic processor.
The effects of scopolamine,desoxyephedrine,amphetamine,caffeine, morphine,reserpine,chlorpromazine,pontobarbital and meprobamate,and their combined effects on the spontaneous activity of albino mice were determined by the photocell activity cage method.
The carpet factory ofYanshan petrochemical company introduced photoelectric jacquard weave control system of high-low loop pile set from American TUFTCO Company in the beginning of the 1980s, the system exists lots of disadvantages, such as oversize control cabinet's volume, complex circuitries, the quantity of photoelectric cell and otherwise element devices being excessive, easily aging and so on. In addition, it also has the defects that are pattern fabrication process being complex, period being long, expenses being considerable and malfunction frequency, etc.
The following are the major research work:First, both the manual and photoelectric cell counting methods are analyzed, research work related to detect technology based on the computer vision is briefly introduced, and the applications of this technology to automatic count the steel rods are summarized.
With laser and photoelectric tube as laser photoelectric sensor, when a workpiece passes through the light area the light will be blocked by the workpiece and the photoelectric tube will emit a pulse signal, the counting for the passing workpiece just is the workpiece number.
A prototype Cerenkov counter, consisting of a R3432-01(similar to R5505) phototube and a quartz radiator with dimensions of 25 × 30 × 12 mm, was tested in a beam of pions and protons with a 1.19-GeV/c-momentum.
An organic scintillator and a phototube with a fine-mesh dynode system are used in the sensor.
The radiation hardness of scintillators and photomultiplier tubes and the phototube stability under long-term optic loading are analyzed.
The detection screen, the phototube and the amplifier are combined for measuring the probe-hole current.
The drum revolution of a motor-driven mycroscope micrometric eyepiece is read by means of a phototube arrangement originating calibrated electric pulses.
A new PTEC with a two-stage photoelectric unit, which offers a higher efficiency as compared to a device with a single-stage photocell, is developed.
A new concept of the photothermoelectric converter (PTEC) is proposed, which is free of the main disadvantages (overheating of the photocell and insufficient heat flux in the thermocell) of the existing schemes.
The experimental results show that the photocell can generate large and stable photocurrent and photovoltage, the photoelectric device made from TC16PyP LB films has a potential application value.
Ants intercept a light beam as they pass under a gantry between the light source and a photocell.
Ten layers of oriented bacteriorhodopsin (BR) LB films are prepared on ITO conductive glass to constitute an ITO electrode/BR LB films/KCl solution/Cu electrode wet photocell.